See also: What are A§E tests?
- Select Loudness Scaling in the ribbon and click on New Measurement
- Select the machine and the test frequency (250, 500, 1000, 2000, 4000, or 6000 Hz) from the drop-down list
- You can choose to check the Run Short Test box to reduce the number of presentation levels and save time
Read also our white paper on the validation of the Short version of this test...
- Use the controls to select the side of the test ear and the conduction
- In the Training mode, you can set a presentation level and click Present. Alternatively, you can check the Auto Scan box and click Scan to make Audiqueen present all intensities from soft to loud.
- In the Test mode, Audiqueen presents the stimulus at different presentation levels and the patient should say how loud this feels.
- The Repeat button can be used to repeat the previous sound once more.
- After the last presentation, Audiqueen calculates the median scores per presentation level and a global score expressed as an RMS value.
Audiqueen calculates the RMS value as a measure of the error compared to the normal line. This RMS (root mean square) is the root of the sum of (the median response – the normal response) for all intensities (presentation levels). As such, this is a positive value. Audiqueen adds a sign mark. This is the sign mark of the sum of all responses – the sum of all normal responses.
You may also have a look at the following videos on loudness scaling:
See also: How to perform A§E selftests?
|This feature is not available in the web viewer.|